Neonatal transport incubator nice 3000
Microprocessor Controlled System with Air and Skin Modes of operation.
Displays Air / Skin Temperature simultaneously with Set temperature
Feather Touch operation with large digital display and comprehensive alarms
Memory backup to restore set temperature and control mode automatically in the event of power failure
An integral and withdrawable baby tray to aid resuscitation during transport Internal battery Backup for 4 hours minimum operates both on AC and DC currents, with visual indication for battery power status and power sources
Collapsible stainless steel stands with shock absorbers
Dual Thermistor Temperature Probes, with accuracy of +0.2°C, to minimize down time and ensure accuracy. Probe interchangeable without calibration
A, 2D (or) 2E tank mounting facility to mount gas cylinders
Unimpeded access to the patient with frontal access and a minimum of 2 access ports and three elbow operated ports
Easy to clean and disinfect
High Intensity CFL Light for observation
Comprehensive alarms: Air Temp. Low / High, Skin Temp. Low / High, Sensor Failure, Low Battery, Heater Failure, Power Failure, Air Flow Failure, Temperature > 39 ° C
Automatic Heater cut off if air / skin temperature inside the incubator exceeds 39° C
Accessories: Auxiliary Tray for placing additional equipment
Optional Accessories: Pressure Regulator, Flow meter, O2 Cylinder, , O2 Driven Venturi Slow suction.
The ‘Phototherapy Apparatus’ research study covers each and every aspect of the Phototherapy Apparatus market globally, which starts from the definition of the Phototherapy Apparatus industry and develops towards Phototherapy Apparatus market segmentations. Further, every segment of the Phototherapy Apparatus market is classified and analyzed on the basis of product types, application, and the end-use industries of the Phototherapy Apparatus market. The geographical segmentation of the Phototherapy Apparatus industry has also been covered at length in this report.
Nice Neotech Medical Systems Pvt. Ltd. (IN) – LED Phototherapy
Certification of CE Under Medical equipments MDD 93/42/EEC for series of Phototherapy Unit by UL International (UK) Ltd. on March 2011
Nine Hi power LED provides to cover the bed area and for effective Phototherapy treatment No ultraviolet rays ensures minimize the UV exposures
No infrared rays ensures minimize the water loss
Adjustable Height – Conveniently allows you to control irradiance levels on the neonate and different types of cradle.
Convenient Tilting of Source Unit, Height Adjustment with Single hand and Castor with elegant maneuvering helps to place the system at different positions, at user convenience.
Treatment Timer ensures the auto cutoff when elapse the set time
Lamp usage hour monitor, Lamp replacement alert are ensures the less downtime service. Lamp usages hours greater than 20,000 hours ensures no need of frequent change of lamp.